Ronald S. Pennock, M.D.; Leonard S. Dreifus, M.D.; Yoshio Watanabe, M.D.; Dryden Morse, M.D.
This content is PDF only. Please click on the PDF icon to access.
At present, permanent cardiac pacemakers are electively replaced after 18 months. Some generators have failed earlier, and thus far no foolproof system can predict impending failure. A new system was developed to monitor the pulse interval to ±0.1 msec, utilizing the electronic R-R signal that is transmitted over ordinary telephone lines by a data coupler to the pacemaker clinic. After pacemaker implantation a base line is established over a 3-week period. The unit is tested monthly for 1 year. Pacemakers with an interval drift of more than ±20 msec in 1 month are then tested weekly until predicted failure (±15
Pennock RS, Dreifus LS, Watanabe Y, Morse D. Comprehensive Evaluation of Cardiac Pacemakers.. Ann Intern Med. ;74:820–821. doi: 10.7326/0003-4819-74-5-820_6
Download citation file:
Published: Ann Intern Med. 1971;74(5):820-821.
Copyright © 2018 American College of Physicians. All Rights Reserved.
Print ISSN: 0003-4819 | Online ISSN: 1539-3704
Conditions of Use