David L. Hayes, MD; Ronald E. Vlietstra, MB, ChB
Hayes DL, Vlietstra RE. Pacemaker Malfunction. Ann Intern Med. 1993;119:828-835. doi: 10.7326/0003-4819-119-8-199310150-00009
Download citation file:
Published: Ann Intern Med. 1993;119(8):828-835.
The field of cardiac pacing has expanded rapidly in recent years. Engineering improvements and microprocessor technology have resulted in a vast increase in pacemaker technology. The high-tech edge of cardiac pacing often discourages all but the pacemaker specialist from approaching a malfunctioning pacemaker. Electrocardiographic signs of pacemaker malfunction can be grouped into four categories: failure to output, failure to capture, undersensing, and inappropriate pacemaker rate. For each of these categories, there may be true malfunctions and pseudomalfunctions. In addition, environmental sources of electromagnetic interference, both within and outside the hospital environment, can result in pacemaker malfunction. Approaching pacemaker malfunction with these categories in mind should help minimize confusion.
Learn more about subscription options.
Register Now for a free account.
Cardiology, Rhythm Disorders and Devices.
Results provided by:
Copyright © 2017 American College of Physicians. All Rights Reserved.
Print ISSN: 0003-4819 | Online ISSN: 1539-3704
Conditions of Use
This PDF is available to Subscribers Only