JOHN OWEN, M.D.; EREN PAYNE, R.T.; KELVIN CARSTAIRS, M.B.
This content is PDF only. Please click on the PDF icon to access.
To the editor: We wish to draw attention to the problem of erroneous results of activated partial thromboplastin times (aPTT) when plasma samples from patients on heparin are tested with the Lancer CoagulyzerTm. The errors can be large enough to interfere with appropriate therapy.
The Coagulyzer is an automated clot detector with a turntable capable of holding 60 samples, and it has two automatic reagent pipettes, positioned eight stations apart; six of these are heated to 37 °C. When the machine is set to the aPTT mode, the first pipette delivers the aPTT reagent (in our laboratory, General Diagnostics [Morris
OWEN J, PAYNE E, CARSTAIRS K. Erroneous Activated Partial Thromboplastin Time. Ann Intern Med. 1978;89:146–147. doi: 10.7326/0003-4819-89-1-146_2
Download citation file:
Published: Ann Intern Med. 1978;89(1):146-147.
Copyright © 2019 American College of Physicians. All Rights Reserved.
Print ISSN: 0003-4819 | Online ISSN: 1539-3704
Conditions of Use